The HS2 is the handheld display for the new Hydrosense II soil-water sensor. When paired with either the CS658 or CS659 portable soil-water probes, it replaces the previous Hydrosense soil-water system. It features a new handheld display and a new probe design. The HS2 handheld has been designed to be compact and portable, with the layout of the buttons allowing for operation with one hand.Read More
The display consists of a three-inch LCD and four navigation buttons that make changing settings and taking measurements as easy as possible. An integrated GPS tags each measurement with a latitude and longitude.
Zones can be created on the unit which group measurements together so that average soil moisture can be calculated for an area. The current position and zone are shown on the display so that measurements can be taken in the same zone.
Data storage has been added to allow more than 1000 measurements to be stored on the device. The data can then be downloaded to a PC via Bluetooth for viewing and archiving.
The new soil moisture probe uses the same accurate measurement technique as the old probe, but the probe housing has been redesigned to aid insertion into and removal from hard soils. The probe rods are secured to the probe housing with ferrule nuts to provide extra stability during insertion. A molded plastic grip connects the sensor cable to housing to provide better grip.
New PC software has been developed to make the most of the data storage capability of the HS2. The software connects to the HS2 via Bluetooth to avoid the need for extra cables.
The software allows the user to:
|Measurements Made||Volumetric water content of porous media (such as soil)|
|Required Equipment||HS2 is a complete system.|
|Soil Suitability||Short rods are easy to install in hard soil. Suitable for soils with higher electrical conductivity.|
|Sensors||Interchangeable sensors; can swap the 12 cm and 20 cm sensors with the reader.|
|Display||128 x 64 pixel graphic LCD|
|Backlight||Blue and white LED (brightness-adjustable)|
|Bluetooth Range||~10 m (~30 ft)|
|Data Storage||> 1000 records (ring memory)|
|Zone Storage||> 100 records|
|Power Supply||6 Vdc, 4 AA batteries|
|Battery Life||6 to 12 months (typical usage)|
|Dimensions||200 x 100 x 58 mm (7.9 x 3.9 x 2.3 in.)|
|Weight||340 g (12 oz)|
Typical Power Consumption of Display
|Backlight Off||2 mA|
|Backlight at 60%||18 mA|
|Backlight at 100%||30 mA|
|GPS Active||35 mA|
|Bluetooth Active||30 mA|
|-NOTE-||The CS659 and CS658 cannot share rods.|
|Water Content Accuracy||
|Volumetric Water Content Resolution||< 0.05%|
|Volumetric Water Content Range||0% to 50% VWC|
|Cable Length||250 cm (98 in.) extended|
|Rod Diameter||5 mm (0.14 in.)|
|Body Dimensions||100 x 92 x 40 mm (3.9 x 3.6 x 1.6 in.)|
|Weight||450 g (15.9 oz)|
The HS2 is the display used in the Hydrosense II System.
The water content sensor is ordered as an option, and may also be purchased separately. Either the CS658 or CS659 can be used with the HS2. The CS658 has 20-cm rods, and the CS659 has 12-cm rods.
Note: The CS658 and CS659 cannot share rods (i.e., 12-cm rods cannot be used with the CS658, and 20-cm rods cannot be used with the CS659).
Current HS2 (Hydrosense II) firmware.
Current version of the HydroSense II Support Software.
Number of FAQs related to HS2: 18
Expand AllCollapse All
When Deficit Mode is used, a negative RWC value indicates that the water content is below the dry reference. For more information, see the "Water Deficit Data" subsection in the "Operation" section of the HS2 Instruction Manual.
No. The HS2 uses Bluetooth communication.
A VWC reading of (---) indicates either that the water content of the soil is greater than 50% or that the EC (electrical conductivity) of the soil is out of range. For more information, see Sections 6.3, 188.8.131.52, and 184.108.40.206 of the HS2 Instruction Manual.
Yes. See the "Connecting via Bluetooth" subsection in the "Data Storage and Retrieval" section of the HS2 Instruction Manual.
When Deficit Mode is used, a value greater than 100% RWC indicates that the water content is above the wet reference. For more information, see the "Water Deficit Data" subsection in the "Operation" section of the HS2 Instruction Manual.
No. The displays and sensors are not interchangeable between the HydroSense and HydroSense II.
No. The original HydroSense was not capable of storing data. The HS2 produces a CSV file that can be merged into an existing data file used for HydroSense data. However, the CSV output file from the HS2 will include additional parameters that were not measured by the original HydroSense.